[R(OHM)-T(C) calibration]
FitMethod = Polynom
TransformX = Line
TransformY = RELPT100
Power = 1
Center = 0
Scale = 1
Fixed = 0
Coeff[0] = 0.00068409711609469
Coeff[1] = 0.991426124511416
Bounds = 18, 393.8574892
Data R(OHM)       T(C)  Weight
     18.68024331  -200  1     
     100.8647     0     1     
     393.8574892  850   1     
End Data
Notice Text
T1 Pt-100 calibration
End Notice Text
